Software update MORA-Nova – Update 22.214.171.124
Modification: During the substance search, the mode button for A / A-Bar for substances that have not yet been marked remains in the state last set. This prevents a constant change of substances that you want to test later in the A-Bar mode.This setting is automatically reset when the substance search window is closed.
Modification: If in the Cornelissen module, a substance is placed directly in a honeycomb without confirmation or evaluation, it automatically receives the current standard amplification and the selected mode. This is indicated when the data is transferred to the satellite software and on the printout.
Modification: The substance test module has a new system for faster work with substance testing. Transmission is prevented while moving the touchscreen. When moving forwards or backward using the button on the measuring pen, a delay time of 300 ms is set, in which it is possible to move without substance transfer.
Modification: The Cornelissen module has a new system for faster work with substance testing. Transmission is prevented while moving the touchscreen. When moving forwards or backward using the button on the measuring pen, a delay time of 300 ms is set, in which it is possible to move without substance transfer.
Modification: In the service dialog, the key areas have been enlarged to make it easier to enter the password or a service code.
Modification: In the modules, CORNELISSEN and SUBSTANCE TEST the display of the transfer status (0% or 100%) has been accelerated and now shows the progress. If the transfer by substance selection is not active or no substance has yet been selected for transfer, the transfer indicator is shown in yellow.
Network: If several NOVA devices are operated in the network, it must be ensured that all devices are on the same (current) version. If a substance is selected in the substance search on a device that is not yet available on the current device, the heading in the substance search is displayed in red. This indicates that there is a newer version of the NOVA device in the network and that it has selected substances from a test set that has not yet been installed (test set definition is still missing on the current device). This function is only applicable in a network in which different NOVA devices access a central database.
New test set “Be-Life” added (72 substances)
New test set “Medimix 2021” inserted (98 substances)
Update of the “Covid-19” test set. All available vaccines included (Biontech, Moderna, Janssen, AstraZeneca) Update “5 elements” test set. New group of causal testing added
Stability: If the database server aborts in the network, the NOVA automatically tries to reconnect to the database server after 15 seconds. During this time, the title line in the main menu lights up red. This enables you to search for the cause of the problem and prevent data loss from the measurement. If this is not possible, a message is issued. After eliminating the problem, the user can reestablish the connection.
If the database server is disconnected and you have already carried out a test/measurement, the data are retained until you switch off the NOVA or you deny the corresponding storage after the connection has been re-established.
If the database server is aborted during therapy and the connection cannot be automatically re-established within 15 seconds, the therapy will stop after the current therapy step has been completed. You then have the option of returning to the main menu and re-establishing the connection or switching to the internal database server to reload the therapy and continue from the current step. Logging is then not possible because all user data is deleted from the memory when the database connection is changed.
NOVA- Database server update
Changes were made to the NOVA database server to increase speed and network stability. The new version number is: 3.2.0
The Nova database server 3.2.0 has a refined logging system in order to be able to carry out a better root cause analysis in the event of network problems
The Nova database server 3.2.0 now also logs network interruptions, i.e. problems caused by unstable cabling are logged.
Update of the satellite software 2.6 (suitable for the NOVA user interface 3.6)